I just updated to the last firmware and two minutes later the teletype just hang..
this is the simple scene running:
CV 1 N RAND 12
CV.SLEW 1 RAND 50
PROB B : 75
Script one is run by the output of a sine lfo, script two by the sub of the oscillator controlled by script one.
Basically the trigger outs sort of continue to work, its the cv that hangs..
btw i notice that it never hangs when not using external triggering..